|Other Options Include:
additional connector of the same type may be ordered. This provides the
user the ability to change out or replace connectors during development
phases of a project. Once the device's connector has been removed and
replaced with the same or different connector, the device's calibration
and match are not guaranteed.
|• Add male
connector, option 20M
|• Add female
connector, option 20F
|A connector can be
changed in about 5 minutes.
||Add recorder out,
option 001. This
option adds an optional output voltage that is proportional to the
input power. The recorder output voltage ranges from zero to one volt
and is proportional to input power level. The range of power levels
represented by the voltage output is set by the user. The output is
provided through the SMB connector that is located in the same position
as the trigger in.
|If the unit is ordered
with both external
hardware triggering and the recorder out options, the SMB connector is
shared between the recorder out and trigger in functions. Trigger in
and recorder out functions are mutually exclusive functions. As a
result, when the unit's recorder output is turned on the trigger input
option 003. Adds an external hardware input trigger and external
hardware output trigger. Both trigger in and trigger out are SMB
connectors. If the recorder out option is also present, the trigger in
SMB connector is shared between trigger in and recorder out (see "Add
recorder out"). As a result, when the unit's trigger input is turned on
the recorder output is disabled.
filters, option 004. These filters are intended for pulse profiling.
The LB480A comes standard with a 100 kHz video filter. Option 004
offers additional user
selectable hardware filters: 200 kHz, 300 kHz, 500 kHz, 1 MHz, 2 MHz, 3
MHz, 5 MHz, and 10 MHz.
Add real-time, calibrated detector output capability with option OW2.
This capability allows viewing of one-shot pulses and other real-time
signals by using a high speed oscilloscope to display the detector's
analog output. The option OW2 signal is available on the TO SMB
connector (trigger out signal
will not be available). Option OW2 requires option 004 and is not
available when option 001 is ordered.